In honor of Tektronix's 65th anniversary, which they celebrated this past week, here's another great old scope. The Tektronix 511 was the first Tektronix product, introduced in 1947. This scope is a 511A, which was introduced a few years later.
I actually bought this scope (serial number 2520) in California in January 2011 with the hope of showing it off to Jim (and perhaps working with him to restore it). Unfortunately, he passed away before I even got a chance to tell him about it.
The purchase was almost a nightmare. Almost. I bought this scope from a guy on Craigslist. He didn't know anything about it; he had found it (and quite a bit of old ham gear) in the basement of his new house when he moved in, and he was trying to sell it all for some free space and a few bucks. (Apparently, the previous owner had passed away, and his kids just sold the house with everything in it.)
When I arrived to purchase it, he had it in the kitchen, set up and plugged in. He said, "I've tested it and it shows a trace on the screen," and he flipped on the power switch. My heart stopped! Conventional wisdom is that these old scopes require a careful power-on ritual after being stored for a long time: the electrolytic capacitors go bad, the high-voltage section goes bad, and dirt and grime get into bad places. Many aficionados recommend cleaning the scope, checking all the big capacitors (especially the "black beauties of death"), and bringing up the power slowly with a variac and an ammeter. This guy just flipped the switch (as much as I wanted, I figured screaming "Oh, God, no!" would have been rude). Thankfully, there was no smoke, and after the trace appeared, I quickly turned the scope off, and said "I'll take it."
I wonder how many old scopes have been damaged by over-enthusiastic sellers trying to demonstrate that "it works" to a buyer?